Imaging Systems: RP1 Imaging System
|Size||3” (W) x 3” (D) x 8.75” (H); 0.05 ft3 (1,290 cm3)|
|Ergonomics||Easy grip imaging head; dovertail-mounted cage; 10’ black silicone umbilical cords|
|Device coupling||Locking and rotating attachment of any device with standard-sized eyepiece|
|Air flow||None. Passive cooling|
|Storage Conditions||50° to 100° F (10° to 38° C); 20%-75% relative humidity, non-condensing|
|50° to 100° F (10° to 38° C); 20%-75% relative humidity, non-condensing
|FLARE® compatibility||RP2 and RP1 photonic control units (PCUs)|
|Positioning||Handheld or locking dovetail mount to accessory arm via cage|
|Light guide||Custom right-angle, 5 mm diameter medical grade light guide with high VIS-NIR transmission|
|Operational verification||On-demand using PCU calibration system|
|Laser safety||Laser-on LEDs 180˚ apart on imaging head|
|Optional sterile drapes for 5 mm and 10 mm diameter eyepiece-coupled devices (one-time use)
|Reach||10’, black silicone umbilical cord; 10’, black silicone VIS/NIR light guide|
|Working distance||Variable, depending on eyepiece-coupled device|
|Eyepiece coupler||Independent adjustment of zoom, focus, and aperture|
|Simultaneous video streams||4 (color video, NIR Channel #1, NIR Channel #2, Color- NIR Merge|
|Color video channel||400-660 nm illumination w/ variable intensity (see PCU); 400-650 nm imaging|
|NIR channel #1||Illumination Class 3R laser, 665 ± 3 nm w/ variable intensity (see PCU); Imaging 685-735 nm|
|NIR channel #2||Illumination Class 3R laser, 760 ± 3 nm w/ variable intensity (see PCU), Imaging ≥ 781 nm|
|Merge channel||Geometric or mathematical, including dual-NIR ratiometric|
|Image resolution||1024 x 1024 pixels|
|Bit depth||10-bits each for red, green, blue, NIR Channel #1, and NIR Channel #2|
|Exposure time||66 µsec to 3 sec; each channel with independent exposure|
|Readout rates||20 Hz at full resolution of all channels|
CURADEL will provide the following service and support for FLARE Devices during the term of the applicable warranty.
Customers of Curadel FLARE® devices may opt to purchase Curadel Platinum Service, a comprehensive maintenance and service contract that will cover each purchased device for an additional 12 (twelve) months beyond the expiration of the warranty. All maintenance and repairs on devices not under an active warranty or service contract in need of repair will be billed at the standard labor rate plus cost of parts.
The above warranties do not cover defects resulting from improper or inadequate maintenance, installation, repair or calibration performed by Customer or a third party not authorized by CURADEL; Customer or third party supplied hardware or software, interfacing or supplies; unauthorized modification; improper use or operation outside of the Specification for the Product; abuse, negligence, accident, loss or damage in transit; or improper site preparation.
Necessary cookies are absolutely essential for the website to function properly. This category only includes cookies that ensures basic functionalities and security features of the website. These cookies do not store any personal information.
Any cookies that may not be particularly necessary for the website to function and is used specifically to collect user personal data via analytics, ads, other embedded contents are termed as non-necessary cookies. It is mandatory to procure user consent prior to running these cookies on your website.